Finge Clip Pulse Oximeter
|FOB Price:||Get Latest Price|
|Min. Order:||10 Pieces|
|Production Capacity:||5000PCS Per Month|
|Transport Package:||Standard Carton|
|Payment Terms:||L/C, T/T, Western Union|
- Model NO.: JAX-204
- Specification: CE&ISO
- Trademark: longfian
- Origin: China
1. Operation of the product is simple ang convenient.
2. The product is small in volume, light in weight(total weight is about 50g including batteries)and convenient in carrying.
3. Power consumption of the product is low and the two originally-equipped two AAA batteries can be operated continuously for 30 hour.
4. Low voltage warning will be indicated in visual window when battery voltage is so low that normal operation the Oximeter might be inflluenced.
5. The product will automatically be powered off when no signal is in the product for more than 8 seconds.
Product Operation Scope
The fingertip Oximeter can be used to measure human Hemoglobin Saturation and heat through finger. The product is suitable for use in family, hospital(including clinical use in internist/surgery, Anesthesia, pediatrics, intensive care and etc), Oxygen Bar, social medical organizations, physical care in sports(it can be used before or after sports. Continually monitoring is not recommended).
Display Format: LED
SpO2: 35-99% in two-digit 7-segment digital
Pulse Ratio: 30-240 BPM in 3-digit 7-segment digital
Pulse Wave: 6-segment bargraph
Battery Voltage Low Indication: LED signals
Power Requirements: Two AAA size 1.5V Alkali Battery
Power Consumption: Less than 40mA
Resolution: ± 1% for SP02 and ± 1 BPM for Pulse Ratio
Measurement Accuracy: ± 2% in stage of 70%-99% SP02 and meaningless when stage being smaller than 70%. ± 2 BPM for Pulse Ratio
Measurement Performance in Weak Filling Condition: Oxyhemoglobin and pulse value can be shown correctly when pulse-filling ratio is 0.5%
Interference Resistance Capacity against Ambient Light: Deviation is smaller thasn ± 1% between value of Oxyhemoglobin measured in natural lighting indoor condition and present lighting sources and that measured in dark room
Automatically powered off: No finger is in the Oximeter more than 8 seconds.
Oxygen saturation (SpO2)
Patient range: Adult and child
Measurment Range: 35%~100%
Accuracy: & plusmn; 2%(70%~100%), 0%~69% unspecified
Measurment Range: 30 bpm ~250 bpm
Accuracy: & plusmn; 2 bpm
Input: Body surface thermal-sensitive resistor temperature sensor
Measurment Range: 0& deg; C~50& deg; C
Accuracy: & plusmn; 0.1& deg; C
Resolution: 0.1& deg; C
Mode: Audio, visual
Type: SpO2, PR, TEMP
Storage and Review
36 hours SpO2 PRTEMP trend data with corresponding date and time
LCD Screen: 128& times; 64 DOT matrix homochromy LCD
LED indicator: Red and green Dual-color LED
Working: 0& deg; C ~ 45& deg; C
Transport and storage: -20 oC ~ 60 oC
Working: 15% ~ 95%(non-condensing)
Transport and storage: 10 %~ 95%(non-condensing)
Working: 86 KPa ~ 106 KPa
Transport and storage: 50KPa ~ 106 KPa
Type: 2Ahr/6V Ni/MH rechargeable battery
Operating time: 30 hours
Recharge time: 10 hours
AC power adapter:
Input: AC100~ 240V, 50/60Hz, 0.7A
Output: DC 8V, 800mA
120mm& times; 63mm& times; 32mm